The I-300, I-310 and I-320 wideband current probes were designed to facilitate current measurements on conductors where space is limited. The convenient size and shape make them quick to install over existing conductors that are inconvenient to pass through a fixed-window type probe. The aperture of .220 ” diameter allows for testing cables up to RG-58.
The relationship between the sensed current and output voltage is: Vout = ZT x Isensed, where ZT = transfer impedance
|Model||Frequency Range (3 dB points)||Bandpass (±0.5 dB)||Zt||Current Handling Capability||Output Impedance||Connector|
|I-300||180 kHz to 300 MHz||400 kHz to 200 MHz||5||>10 A pulse||50 Ω||SMA|
|I-310||50 kHz to 200 MHz||100 kHz to 125 MHz||1||>10 A pulse||50 Ω||SMA|
|I-320||200 kHz to 500 MHz||150 kHz to 450 MHz(<2 db)||1||>10 A pulse||50 Ω||SMA|